|Citation:||LIANG Kun, FANG Wei, GU Yangyi, CHEN Dehao, HAO Shuangyu, YANG Zhiqiang, FANG Zhanjun, ZHANG Aimin. Remote Time Transfer and Traceability Method, Device and System[J]. Metrology Science and Technology, 2021, 65(4): 3-13. doi: 10.3969/j.issn.2096-9015.2021.04.01|
To solve the problem of remote transmission of legal time measurements and fill the gap of traceable time standards and time-measuring instruments, the remote time traceability method was studied, and the remote time traceability device NIMDO was developed by disciplining rubidium, cesium, and hydrogen atomic clocks with reference to the atomic time national primary standard (UTC(NIM)) and using the GNSS time and frequency transfer method, and a remote time traceability system was preliminarily constructed based on the device. Through various experimental verifications, the device achieves a high-performance time scale at the remote end and synchronizes with UTC(NIM) in real-time, which is equivalent to reproducing UTC(NIM) at the remote end with a certain time and frequency deviations, and the time deviation of NIMDO from UTC(NIM) is better than ±10 ns and frequency deviation is better than ±1×10−13 in over 90% of the time, and the time deviation stays within ±5 ns in more than 87% of the cases.
国家质量监督检验检疫总局. 时间频率计量器具检定系统表: JJG 2007-2015[S]. 北京: 中国质检出版社, 2015.
Kun L, Felicitas A, Gerard P, et al. Evaluation of BeiDou time transfer over multiple inter-continental baselines towards UTC contribution[J]. Metrologia, 2018, 55(4): 513-525. doi: 10.1088/1681-7575/aac586
Lombardi M. A NIST Disciplined Oscillator[C]. Proceedings of the NCSL International Workshop and Symposium, 2010.
Lombardi M A. A NIST Disciplined Oscillator: Delivering UTC(NIST) to the Calibration Laboratory[J]. NCSLI Measure, 2010, 5(4): 46-54. doi: 10.1080/19315775.2010.11721537
Tomonari Suzuyama, Atsuko Okuda, Masaki Amemiya. Long-term properties evaluations of frequency remote calibration system using GPS[C]. Presentation of ATF2015. Beijing: National Institute of Metrology, 2015.
Defraigne P, Petit G. CGGTTS-Version 2E: an extended standard for GNSS Time Transfer[J]. Metrologia, 2015, 52(6): G1-G1. doi: 10.1088/0026-1394/52/6/G1
Liang K, Zuo F, Pei C, et al. Real-Time Remote Calibration (RTRC) System for Time and Frequency[C]. IFCS-EFTF2013, 2013.
Jiang Z, Petit G. Time Transfer with GPS Satellites All in View[C]. Asia-Pacific Workshop on Time and Frequency. Beijing, 2004: 236–43.
Hamilton Technical Services, “Stable 32 user manual,” Technical Report, ID # 212944, Version 1.50, Mar. 2007, pp. 55-63.
Stanford Research System, “SR620 Universal Time Interval Counter,” Technical Report, Revision 2.7, Feb. 2006, pp. 1-5.
Symmetricom, “5125A Phase Noise Test Set Operations and Maintenance Manual,”Technical Report, DOC05125A, Rev A, Mar. 2009, pp. 60-61.
Symmetricom, “5071A Primary Frequency Standard Operating and Programming Manual,” Technical Report, 05071-90041, May 2001, pp. 6-5.
Kun L, Hang Y, Fei Z, et al. Disciplined oscillator system by UTC(NIM) for remote time and frequency traceability[C]. European Frequency & Time Forum. IEEE, 2015.
梁坤, 顾杨义. 远程时间溯源示范体系-时间公报[EB/OL]. https://www.nmdc.ac.cn/u/cms/www/202012/28145204bfgw.pdf.